- Open Access
Accuracy assessment and error analysis for diameter at breast height measurement of trees obtained using a novel backpack LiDAR system
Forest Ecosystems volume 7, Article number: 33 (2020)
The LiBackpack is a recently developed backpack light detection and ranging (LiDAR) system that combines the flexibility of human walking with the nearby measurement in all directions to provide a novel and efficient approach to LiDAR remote sensing, especially useful for forest structure inventory. However, the measurement accuracy and error sources have not been systematically explored for this system.
In this study, we used the LiBackpack D-50 system to measure the diameter at breast height (DBH) for a Pinus sylvestris tree population in the Saihanba National Forest Park of China, and estimated the accuracy of LiBackpack measurements of DBH based on comparisons with manually measured DBH values in the field. We determined the optimal vertical slice thickness of the point cloud sample for achieving the most stable and accurate LiBackpack measurements of DBH for this tree species, and explored the effects of different factors on the measurement error.
1) A vertical thickness of 30 cm for the point cloud sample slice provided the highest fitting accuracy (adjusted R2 = 0.89, Root Mean Squared Error (RMSE) = 20.85 mm); 2) the point cloud density had a significant negative, logarithmic relationship with measurement error of DBH and it explained 35.1% of the measurement error; 3) the LiBackpack measurements of DBH were generally smaller than the manually measured values, and the corresponding measurement errors increased for larger trees; and 4) by considering the effect of the point cloud density correction, a transitional model can be fitted to approximate field measured DBH using LiBackpack- scanned value with satisfactory accuracy (adjusted R2 = 0.920; RMSE = 14.77 mm), and decrease the predicting error by 29.2%. Our study confirmed the reliability of the novel LiBackpack system in accurate forestry inventory, set up a useful transitional model between scanning data and the traditional manual-measured data specifically for P. sylvestris, and implied the applicable substitution of this new approach for more species, with necessary parameter calibration.
Forest structures are generally characterized using metrics such as diameter at breast height (DBH), tree height, and tree density (Dubayah and Drake 2000). The features of Forest structure is normally quantified to reflect the community dynamics and effects of disturbances (Dubayah et al. 2010; Filippelli et al. 2019), to estimate the community biomass and carbon pool (Fang et al. 2001; Ni-Meister et al. 2010), and to indicate the mechanism of community assembly (Allié et al. 2015). Forest structure is also the critical information used in forest management and planning (Wulder et al. 2009).
DBH is one of the most important metrics of forest structure, generally used to indicate age structure or to reflect the radial growth of trees (Muller-Landau et al. 2006). The traditional forestry inventory uses a ruler and rangefinder to measure structural indices such as the DBH and height stem by stem at the forest stand scale (Liu et al. 2018a, 2018b), and predictive models are fitted for regional estimates of forest metrics such as the timber volume or biomass (le Maire et al. 2011). This approach is always limited by the available labor force and operating time. Since the 1980s, vegetation coverage information has been obtained by satellite remote sensing and vegetation indicators such as the normalized difference vegetation index (NDVI) have been designed for estimating the forest vegetation biomass across space with much higher efficiency (Raynolds et al. 2006). However, the traditional remote sensing approach cannot directly obtain forest structure information, so global dynamic vegetation models (DGVMs) generally apply plant functional types as spatial units in simulations (Sato 2009; Bachelet et al. 2018). A lack of vegetation structure information leads to large uncertainty in the vegetation inversion (Meir et al. 2017), and supplementing forest structure information substantially improves the accuracy of DGVMs when predicting the vegetation productivity and estimating the responses of vegetation to climate change (Zhu et al. 2016; Garcia-Gonzalo et al. 2017). In recent years, the rapid development of light detection and ranging (LiDAR) has improved the spatial resolution of remote sensing data to the centimeter level. Compared with traditional spectral remote sensing technology, LiDAR is better at extracting the three-dimensional (3D) structural characteristics of vegetation, and thus it is increasingly used in forestry inventory and forest ecology research (Lim et al. 2003; Davies and Asner 2014; Alonzo et al. 2015).
LiDAR can be divided into three categories comprising space-borne LiDAR, airborne LiDAR, e.g., unmanned aerial vehicle (UAV)-borne LiDAR and airborne laser scanning (ALS), and ground-based LiDAR, e.g., backpack- or vehicle-based LiDAR and terrestrial laser scanner (TLS), according to the loading platform. Space-borne and airborne LiDAR are more efficient at measuring the 3D structure of the vegetation canopy at a larger scale, but less effective at obtaining information regarding the understory structure because of the obstructive effect of the forest canopy (Wu et al. 2015; Fu et al. 2018). In contrast, ground-based LiDAR is better at providing detailed information about the understory vegetation (Moskal and Zheng 2012). For example, Liu et al. (2016) measured the DBH increases for trees in forest communities using fixed ground-based LiDAR and achieved a tree identification accuracy of about 81% in natural forest stands. Among the various categories of ground-based LiDAR, TLS was developed first and it has a high measurement accuracy but the fixed measurement method limits its spatial flexibility, while the capacity of vehicle-based LiDAR is often limited by complex terrain and available roads (Yu et al. 2015). Backpack LiDAR (e.g., LiBackpack) is a novel type of portable LiDAR for which the surveyor is the loading platform, and thus it has a high capacity in terms of accessibility and route choice. Compared with TLS, backpack LiDAR is generally much lighter and more portable, and it can obtain much higher quality 3D point clouds in forest with different vegetation structures (Su et al. 2018). However, the LiBackpack is loaded on the walking surveyors during the operating process, which may significantly reduce the system stability and increase the uncertainty of the measurements.
Studies have assessed the accuracy of the DBH measurements obtained with backpack LiDAR (Holmgren et al. 2017; Oveland et al. 2017, 2018), but the factors that might influence these measurements and their contributions to the error of backpack LiDAR measurements have not been explored. According to our field experience using backpack LiDAR for measuring forest structures, the uncertainty in the results may have the following sources: 1) the effects of irregular LiDAR movements on the variation of the point cloud density; 2) the effects of 3D point cloud sampling method on parameter estimates; and 3) the geometric features of the measured objects, such as size, shape and dip angle of a tree trunk. In order to quantify the potential impacts of these factors on the uncertainty of the forest measurements obtained with this novel instrument, we investigated a Pinus sylvestris var. mongolica plantation containing trees of different ages in the Saihanba Natural Forest Park, Hebei Province in northern China, where we focused on the accuracy and uncertainty of the DBH measurements, the most important forest structure parameter.
Materials and methods
The study was conducted in the Qiancengban Forest District (42.38°–42.48° N, 117.08°– 117.43° E, 1431 m a.s.l.) of Saihanba National Forest Farm, Hebei Province, China. This forest district is located in a mountainous area at the southeastern edge of the Inner Mongolia Plateau, and it has a semiarid temperate climate (Xing et al. 2017). According to the observational data acquired by the local meteorological station in this region from 1960 to 2017, the annual average temperature was − 1.03 °C and the average annual precipitation was 456.87 mm. The main vegetation types comprise artificial coniferous forests planted from the 1960s to the 1980s. The dominant tree species include Pinus sylvestris var. mongolica, Larix gmelinii, and Picea meyeri, as well as scattered natural secondary deciduous broad-leaved forests of Betula platyphylla, and Ulmus pumila woodland. Herbs and shrubs are sparse under the forest canopy.
The LiBackpack LiDAR system used in this study was developed in 2018 by the Beijing Green Valley Technology Co. Ltd. The core parts of the LiBackpack system comprise the VLP-16 Lidar sensor and LiDAR360 software. The VLP-16 Lidar sensor was produced by Velodyne Lidar, Inc., and the technique parameters for the LiBackpack are given in Table S1. A spatial analysis module is embedded in the LiDAR360 software that provides a set of functions on LiDAR point cloud data processing and analyses.
Five plots of Pinus sylvestris var. mongolica forest measuring 30 m × 30 m area were selected. First, we used lining ropes to enclose a square of 30 m × 30 m in the forest, then measured the x and y coordinates of each tree with a DBH > 2.5 cm using a laser range finder (DISTO X3), and manually measured the tree DBH (i.e. DBHfield) using a measuring tape. Meanwhile, an investigator carried a LiBackpack to measure the trees within the sample plots on foot according to the route shown in Fig. S1. From a corner of the plot, the investigator walked along a zigzag line with a row distance of about 5 m, and passed two sides of each tree in the sample plot. During data acquisition, the equipment was carried on the investigator’s back, where the sensor was higher than the top of the investigator’s head and 3D data were scanned automatically. The built-in microcomputer system integrated the relative position and inertial navigation system information to produce 3D point cloud data. Meanwhile, the DBH values were measured for the trees with a measuring tape as field reference data (DBHfield). Basic information of the plots is listed in Table 1.
LiBackpack data processing
Ground points extraction and tree point cloud normalization
For the data file of the scanned point cloud of each sampling plot, the first step of processing was to clean the data and separate the point cloud of the ground from that of the trees, since all objects refluxed the lasers generate groups of points in the 3D point cloud. Ground points were extracted by applying an improved progressive triangulated irregular network densification algorithm (Zhao et al. 2016), and the elevations of ground points were then subtracted from the elevations of the nearest non-ground points. Thus, the locations of all trees were transformed onto a horizontal plane of the same elevation in a data normalization process. The ground points were extracted and gridded to generate a file of digital elevation model (DEM) using an inverse distance weighted interpolation method, with a resolution of 0.5 m. These processes were implemented using the LiDAR 360 V2.2 software (https://greenvalleyintl.com/software/lidar360/).
Point cloud slicing processing
DBH fitting was implemented based on a specific height of the tree stem (Mendez et al. 2014), where the normalized point cloud was sliced horizontally with a specific vertical thickness (recorded as ST; Fig. S2) and the point cloud slices with elevations between 1.3 ± 0.5 H were intercepted subsequently. In order to analyze the effect of the slice thickness on the DBH fitting accuracy, 16 H-values ranging from 5 to 80 cm were specified at intervals of 5 cm. The slicing process was implemented using the LiDAR 360 V2.2 software.
Tree branches can interfere with the effectiveness of the DBH fitting algorithm used for stem recognition (Liu et al. 2018a, 2018b), so it is necessary to cut off the point cloud of branches and leaves so as to obtain a point cloud slices of mere stems. For this purpose, we designed a mask extraction procedure to slice the point clouds (see Supplementary Methods 1 and Fig. S3) using ArcGIS 10.2 and a Python (v.2.7.3) script (Supplementary Method 2).
An adaptive cylindrical fitting method was used to calculate the DBH for slices of the point clouds with different thicknesses. First, we used a density clustering algorithm called density-based spatial clustering of applications with noise (DBSCAN) to automatically divide the sufficiently dense point clouds into different clusters to obtain single tree stem segmentation (Tao et al. 2015). For each cluster, a cylindrical fitting method was applied on the base of 3D Ordinary Least Square (OLS) to obtain the LiDAR DBH values (DBHLi values) and the relative coordinates of each tree stem. Then we used artificial visual interpretation to remove the error fitting. This procedure was conducted using LiDAR 360 V2.2 software. After that, we acquired points in shape file format with DBH and relative coordinates information of each stem. By visual check of point patterns of the scanned data and the field measured data and ranking them by the same order, it was easy to match the records of two DBH data by each stem.
According to the measurement process, we defined four possible sources of error in the LiBackpack measurements: two characteristics of the measured objects comprising 1) DBHfield and 2) the tangent of the stem angle (TSA); 3) an environmental factor comprising the topographic slope (TS); and 4) a data factor comprising the point density (PD). Based on the DEM, we obtained TS using the surface slope tool in ArcGIS 10.2. OLS circle fitting (Thomas and Chan 1989), which was used to locate the centers of top and bottom surfaces of the stem slices, before deriving the axes of the stem cylinders to calculate TSA. PD was calculated as the points on the unit horizontal projection area using Eq. (1):
where N denotes the number of points, i is the order of individual trees, ST is the slice thickness, and S is the horizontal projection area of the stem point cloud.
The DBHLi values were compared with the DBHfield values. The accuracies of the measurements in different slice thicknesses were evaluated based on adjusted R2, root Mean Squared Error (RMSE), relative root mean squared error (rRMSE) using Eq. (2), and the relative accuracy (rA) with Eq. (3):
where H is the slice thickness and j denotes the DBH class. We divided all of the trees into five size classes according to the DBHfield values. We ensured that the sample size in each class was roughly equal and compared rA in each class in different slice thicknesses using analysis of variance (ANOVA).
According to the steps described above, we defined the optimum slice thickness H0 that could obtain the highest rA for each size class, and we analyzed the causes of the measurement errors (ΔDBH) based on H0, where we defined ΔDBH as the dependent variable in Eq. (4) and the four factors as independent variables.
We conducted partial correlation analysis to determine the factors that had significant correlations with ΔDBH, and used these factors to build a multivariate prediction model for correcting the accuracy of the LiBackpack measurements.
Measured versus scanned DBH values for different slice thickness classes
We obtained the DBHfield and DBHLi values for 158 Pinus sylvestris var. mongolica trees. In general, the DBHLi values (157.8 ± 51.0 mm) were significantly smaller than the DBHfield values (169.7 ± 51.2 mm) (t = − 8.3949, p < 0.001). Accord to linear models in Fig. 1, the adjusted R2, RMSE, and fitted slope values varied among different slice thickness classes for the point cloud, where the mean values were 0.85, 2.31 mm, and 0.95, respectively. However, the maximum adjusted R2 and minimum RMSE values were obtained at a slice thickness of 30 cm (Fig. 2). Thus, we selected 30 cm as the optimal slice thickness for DBHLi fitting in the following analyses.
rA values of DBHLi in different tree size classes
For all of the sampled trees pooled into five diameter classes, the rA values for DBHLi varied with the slice thickness. According to the standard deviation of DBHLi for the samples in each diameter class (Fig. 3a), rA was more variable in the smaller than the larger class of DBHfield values. Thus for larger trees, the DBHLi values were more stable with respect to the variation in the slice thickness. ANOVA indicated that rA differed significantly among the DBH classes (p < 0.001), and the mean rA increased as the DBHfield class increased (Fig. 3b).
For the point cloud samples with a slice thickness of 30 cm, the absolute error ΔDBH was negatively correlated with DBHfield (r = − 0.178, p = 0.025), the point cloud density (PD) (r = − 0.496, p < 0.001), and tangent of the stem angle (TSA) (r = − 0.189, p = 0.017). Among the independent variables, PD had a significant positive correlation with DBHfield (r = 0.188, p = 0.029) and TSA (r = 0.349, p < 0.001), and there were no significant correlations among the other variables. A log-transformed linear model fitted for the relationship best between ΔDBH and PD, and predicted a positive ΔDBH only when PD < 4 points·cm− 2 (p < 0.0001). DBHLi was generally smaller than DBHfield at higher PD values (Fig. 4). In addition, a linear decreasing trend was fitted between ΔDBH and DBHfield, thereby indicating that when the tree was larger, the DBH tended to be underestimated to a greater extent by LiBackpack. However, partial correlation tests only found a significant correlation between ΔDBH and PD (r = − 0.466, p < 0.001).
Therefore, based on the point cloud samples with a slice thickness of 30 cm, an OLS transition model was fitted for trees using the DBHfield values and DBHLi values by considering (or not) the point cloud density as a covariate in Eqs. (5) and (6), as follows:
The goodness-of-fit and prediction accuracy were better for model (5), i.e., adjusted R2 = 0.920, RMSE = 14.77 mm, and rRMSE = 0.087, than model (4), i.e., adjusted R2 = 0.890, RMSE = 20.85 mm, and rRMSE = 0.123. The AIC = 1297.7 of model (5) was also much lower than AIC = 1346.4 in model (4). In particular, the RMSE was reduced by 29.2%, which suggested that the model’s predicting capacity could be improved greatly by considering the effect of the point cloud density.
Comparison of DBH measurement among different LiDAR systems
Comparing with the traditional 3D measurement system based on Global Navigation Satellite System and Inertial Navigation System (GNSS+INS) technology, the cost of LiBackpack is lower. Moreover, LiBackpack can implement accurate scanning during its movement and real time data integration, thereby providing the greatest flexibility and high data acquisition efficiency, compared with other forms of LiDAR systems, such as Airborne LiDAR, UAV LiDAR, TLS and Vehicle-based LiDAR (Anderson et al. 2018; Herrero-Huerta et al. 2018; Paris and Bruzzone 2019; Polewski et al. 2019). However, the stability of LiBackpack platform is probably the lowest among these types of LiDAR system, thus the reduction of data acquisition accuracy should be a sacrifice to the flexibility, and this function trade-off is critical for the selection of platforms in practical applications.
In our study, the best prediction model for the LiBackpack measurements obtained results of R2 = 0.920, RMSE = 14.57 mm, and rRMSE = 0.087. Holmgren et al. (2017) also used backpack LiDAR to measure the DBH for trees in four plots with sample sizes ranging from 50 to 90 individuals, where the average RMSE = 18.5 mm and rRMSE = 0.06. Oveland et al. (2017, 2018) used backpack LiDAR to measure the DBH for 18 and 199 trees, where the RMSE values were 22 and 15 mm, respectively, and the rRMSE values were 0.075 and 0.091. We also collected more than 50 LiDAR-based previous reports on measuring experiments in the past 5 years (Table 2), and found that the RMSE values were significantly higher for ALS measurement than TLS measurements (p < 0.0001), but there were no significant differences between the measurements obtained using different TLS scanning modes (p = 0.58) (Fig. 5).
Moreover, the results obtained with different DBH fitting methods did not differ significantly (p = 0.07). The average RMSE values for single and multi-station TLS measurements were 20.1 and 15.5 mm, respectively, with average rRMSE values of 0.078 and 0.082. The RMSE values were larger for TLS measurements than LiBackpack measurements in the present study, but the rRMSE values of TLS measurements were smaller. LiBackpack measurements were clearly better than ALS measurements (RMSEmean = 6.4 mm, rRMSEmean = 0.251). Thus, there is an obvious trade-off between the accuracy and efficiency of LiBackpack DBH measurements, where moving the LiBackpack under the forest canopy ensures that more complete and uniform point clouds are scanned for stems by the equipment, but simplifying the equipment hardware to improve portability tends to yield lower rA values compared with the TLS measurements (Su et al. 2018).
Optimal thickness of point cloud slice for DBH estimation
The DBH was estimated for trees based on the scanned point cloud using the adaptive cylindrical fitting method, so it was essential to determine the optimal thickness for the sample slice in the point cloud. Two main factors could lead to uncertainty in the estimates. First, in order to correct the effect of the stem dip angle, a sample slice should be sufficiently thick to achieve satisfactory accuracy. Second, the uneven stem surface, especially the bumps, cracks, and branches could introduce substantial noise in the estimate, so a sufficiently thick stem sample is necessary to smooth the unevenness of the trunk. However, the sample could be more variable when the slice is thicker. Therefore, the solution involves finding a balance between the two sources of uncertainty. In our experiment on the P. sylvestris var. mongolica population, a thickness of 30 cm was confirmed as the optimal sample slice by multiple standards. However, the generalizability of this value requires further validation in other working contexts or on other species, and the parameter calibration of this simple model is still critical in its application.
Sources of uncertainty in LiBackpack DBH measurements
The estimated DBHLi values were obviously correlated with the DBHfield values, but our experiment indicated that the LiBackpack scanned DBH values were generally smaller than the manually measured values, and the absolute error ∆DBH increased with the tree size (p = 0.025, Fig. 4b). Moreover, ∆DBH had a log-transformed negative correlation with the point cloud density (p < 0.001), and DBHLi was larger than DBHfield only when the tree corresponded to a thin point cloud (PD < 4 points·cm− 2). In addition, the dip angle of a tree and the topography slope had weak but significant associations with ∆DBH, mainly due to their effects on the scanned point cloud density. Liu et al. (2018a, 2018b) also obtained the maximum rA when using low density point clouds. However, why did the DBH tend to be underestimated to a greater extent for a larger tree, or for a tree estimated using a thicker scanned point cloud? What is the relationship between tree size and the point cloud density?
The variations in the PD could be attributed to the scanning distance, scanning angle, and scanning frequency by sensors (Anderson et al. 2018). In general, a larger tree has a rougher surface and irregular intersectional shape, with deeper grooves and uneven outer skin. Manual measurement involves wrapping a tape around the outermost surface of the tree to determine the maximum girth, whereas the LiBackpack scanner may emit lasers into the grooves and return a point cloud circle with a particular thickness (Johnson 2009), which is wider for a larger tree. A fitting algorithm based on measurements of either a circle or a cylinder determines the DBH based on a circle passing through the locations with the highest density in the stem point cloud (Liu et al. 2018a, 2018b), which are generally in the middle. In contrast, smaller trees usually have more branches, which tend to obscure the access by the laser, thereby resulting in a thinner point cloud circle. Therefore, the differences in the surface structure of larger and smaller trees may explain the negative correlations between ∆DBH and tree size, and with PD.
The negative correlations between ∆DBH and tree size as well as PD can also be due to the shelter and overlap among trees in a community. When scanning a forest of trees with different DBHs, the walking route should be designed to move through the plot as uniformly as possible (Fig. S1), but a larger tree will always be more exposed to the lasers, whereas a smaller tree is more likely to be partially or completely blocked by neighboring trees. Thus, the average laser exposure time will be lower for a smaller tree than a larger tree. Moreover, the tangent of the stem dip was positively correlated with the point cloud density, although the correlation was weak (p = 0.05), and thus a larger angle for a tree might lead to more laser returns per unit area.
In this study, we used DBH measurement obtained with a measuring tape as the field reference data, and these data have been used in most forestry surveys and plant community studies (Srinivasan et al. 2015; Polewski et al. 2019; Yun et al. 2019). In order to integrate new LiDAR-scanned forest structure measurements with the huge amounts of existing historical data, it would be useful to develop a transition model to link these two data sources. In the present study, we quantified and examined the effects of point cloud features on the quality of the data transition, and proposed a suitable multivariate model with substantially improved accuracy. Further experiments will be needed to explore the uncertainty related to trees of different shapes (or species), different measuring environment and different types of LiDAR scanners. The PD at a low and more stable level is critical for obtaining more accurate measurements, so the experimental design needs to be considered, such as the movement path and speed, scanning time, and the effect of the understory structure of trees on the quality of the scanned data. This test should also apply to the confirming test of the optimal slice thickness, although 30 cm thick here for P. sylvestris var. mongolica, for variable measuring situations.
The application of LiDAR in forestry investigations is expected to substantially improve their efficiency, but assessing the accuracy and sources of uncertainty are essential for LiDAR data. In this study, we measured the DBH for 158 Pinus sylvestris var. mongolica trees using the manual method and a backpack LiDAR system called LiBackpack. Compared with samples from all other thickness classes, a slice of the stem point cloud with a vertical thickness of 30 cm obtained the optimal match between DBHLi and DBHField. The DBHLi values were generally smaller than the DBHField values, and the difference was primarily determined by the point cloud density used to the estimate DBHLi. The branches of small trees and the rough surfaces of large trees were the major sources of the uncertainty in the PD. After correction based on PD, the accuracy of the DBH estimates obtained using LiBackpack measurements was similar to that of the TLS measurements. The reasonable data accuracy and high access capacity make LiBackpack an efficient approach for mapping and estimating the structure of forests and woodlands at broad scales.
Availability of data and materials
The datasets used during the current study are available from the corresponding author on reasonable request.
Analysis of variance
Airborne laser scanning
Diameter at breast height
Density-based spatial clustering of applications with noise
Digital elevation model
dynamic vegetation models
Global Navigation Satellite System and Inertial Navigation System
Global positioning system
Light detection and ranging
normalized difference vegetation index
Ordinary least square
Random sample consensus
Root mean squared error
Relative root mean squared error
Simultaneous localization and mapping
Support vector regression
Terrestrial laser scanner
Multi-scan terrestrial laser scanning
Single-scan terrestrial laser scanning
Tangent of the stem angle
Unmanned aerial vehicle
Allié E, Pélissier R, Engel J, Petronelli P, Freycon V, Deblauwe V, Soucémarianadin L, Weigel J, Baraloto C (2015) Pervasive local-scale tree-soil habitat association in a tropical forest community. PLoS One 10(11):e0141488
Alonzo M, Bookhagen B, McFadden JP, Sun A, Roberts DA (2015) Mapping urban forest leaf area index with airborne LiDAR using penetration metrics and allometry. Remote Sens Environ 162:141–153
Anderson KE, Glenn NF, Spaete LP, Shinneman DJ, Pilliod DS, Arkle RS, McIlroy SK, Derryberry DR (2018) Estimating vegetation biomass and cover across large plots in shrub and grass dominated drylands using terrestrial LiDAR and machine learning. Ecol Indic 84:793–802
Bachelet D, Ferschweiler K, Sheehan T, Sleeter B, Zhu Z (2018) Translating MC2 DGVM results into ecosystem services for climate change mitigation and adaptation. Climate 6(1):1
Bu G, Wang P (2016) Adaptive circle-ellipse fitting method for estimating tree diameter based on single terrestrial laser scanning. J Appl Remote Sens 10(2):026040
Davies AB, Asner GP (2014) Advances in animal ecology from 3D-LiDAR ecosystem mapping. Trends Ecol Evol 29(12):681–691
Dubayah RO, Drake JB (2000) LiDAR remote sensing for forestry. J Forest 98(6):44–46
Dubayah RO, Sheldon SL, Clark DB, Hofton MA, Chazdon RL (2010) Estimation of tropical forest height and biomass dynamics using LiDAR remote sensing at La Selva, Costa Rica. J Geophys Res Biogeosci 115(G2):272–281
Fang JY, Chen AP, Peng CH, Zhao SQ, Ci L (2001) Changes in forest biomass carbon storage in China between 1949 and 1998. Science 292(5525):2320–2322
Filippelli SK, Lefsky MA, Rocca ME (2019) Comparison and integration of LiDAR and photogrammetric point clouds for mapping pre-fire forest structure. Remote Sens Environ 224:154–166
Fu L, Liu Q, Sun H, Wang Q, Li Z, Chen E, Pang Y, Song X, Wang G (2018) Development of a system of compatible individual tree diameter and aboveground biomass prediction models using error-in-variable regression and airborne LiDAR data. Remote Sens 10(2):325
Garcia-Gonzalo J, Zubizarreta-Gerendiain A, Kellomäki S, Peltola H (2017) Effects of forest age structure, management and gradual climate change on carbon sequestration and timber production in Finnish boreal forests. In: Bravo F, LeMay V, Jandl R (eds) Managing Forest ecosystems: the challenge of climate change. Springer International Publishing, Switzerland, pp 277–298
Herrero-Huerta M, Lindenbergh R, Rodriguez-Gonzalvez P (2018) Automatic tree parameter extraction by a Mobile LiDAR system in an urban context. PLoS One 13(4):e0196004
Holmgren J, Tulldahl H, Nordlöf J, Nyström M, Olofsson K, Rydell J, Willén E (2017) Estimation of tree position and stem diameter using simultaneous localization and mapping with data from a backpack-mounted laser scanner. Inte Arch Photogramm Remote Sens Spatial Inform Sci 42:25–27
Johnson SE (2009) Effect of target surface orientation on the range precision of laser detection and ranging systems. J Appl Remote Sens 3(1):033564
le Maire G, Marsden C, Nouvellon Y, Grinand C, Hakamada R, Stape J-L, Laclau J-P (2011) MODIS NDVI time-series allow the monitoring of Eucalyptus plantation biomass. Remote Sens Environ 115(10):2613–2625
Lim K, Treitz P, Wulder M, St-Onge B, Flood M (2003) LiDAR remote sensing of forest structure. Prog Phys Geogr 27(1):88–106
Liu C, Xing Y, Duanmu J, Tian X (2018b) Evaluating different methods for estimating diameter at breast height from terrestrial laser scanning. Remote Sensn 10(4):513
Liu G, Wang J, Dong P, Chen Y, Liu Z (2018a) Estimating individual tree height and diameter at breast height (DBH) from terrestrial laser scanning (TLS) data at plot level. Forests 9(7):398
Liu L, Pang Y, Li Z (2016) Individual tree DBH and height estimation using terrestrial laser scanning (TLS) in a subtropical forest. Sci Silv Sin 52:26–37 (in Chinese)
Meir P, Shenkin A, Disney M, Rowland L, Malhi Y, Herold M, da Costa ACL (2017) Plant structure-function relationships and woody tissue respiration: upscaling to forests from laser-derived measurements. In: Ghashghaie J, Tcherkez G, eds. Plant respiration: metabolic fluxes and carbon balance. In: Govindjee, Sharkey TD, eds. Series: Advances in Photosynthesis andRespiration. Dordrecht, Springer 43, 91–108.
Mendez V, Rosell-Polo JR, Sanz R, Escola A, Catalan H (2014) Deciduous tree reconstruction algorithm based on cylinder fitting from mobile terrestrial laser scanned point clouds. Biosyst Eng 124:78–88
Moskal LM, Zheng G (2012) Retrieving forest inventory variables with terrestrial laser scanning (TLS) in urban heterogeneous forest. Remote Sens 4(1):1–20
Muller-Landau HC, Condit RS, Chave J, Thomas SC, Bohlman SA, Bunyavejchewin S, Davies S, Foster R, Gunatilleke S, Gunatilleke N, Harms KE, Hart T, Hubbell SP, Itoh A, Kassim AR, LaFrankie JV, Lee HS, Losos E, Makana JR, Ohkubo T, Sukumar R, Sun IF, Nur Supardi MN, Tan S, Thompson J, Valencia R, Muñoz GV, Wills C, Yamakura T, Chuyong G, Dattaraja HS, Esufali S, Hall P, Hernandez C, Kenfack D, Kiratiprayoon S, Suresh HS, Thomas D, Vallejo MI, Ashton P (2006) Testing metabolic ecology theory for allometric scaling of tree size, growth and mortality in tropical forests. Ecol Lett 9(5):575–588
Ni-Meister W, Lee S, Strahler AH, Woodcock CE, Schaaf C, Yao T, Ranson KJ, Sun G, Blair JB (2010) Assessing general relationships between aboveground biomass and vegetation structure parameters for improved carbon estimate from LiDAR remote sensing. J Geophys Res Biogeosci 115(G2). https://doi.org/10.1029/2009JG000936
Olofsson K, Holmgren J, Olsson H (2014) Tree stem and height measurements using terrestrial laser scanning and the RANSAC algorithm. Remote Sens 6(5):4323–4344
Oveland I, Hauglin M, Giannetti F, Kjorsvik NS, Gobakken T (2018) Comparing three different ground based laser scanning methods for tree stem detection. Remote Sens 10(4):538
Oveland I, Hauglin M, Gobakken T, Naesset E, Maalen-Johansen I (2017) Automatic estimation of tree position and stem diameter using a moving terrestrial laser scanner. Remote Sens 9(4):350
Paris C, Bruzzone L (2019) A growth-model-driven technique for tree stem diameter estimation by using airborne LiDAR data. IEEE Transact Geosci Remote Sens 57(1):76–92
Polewski P, Yao W, Cao L, Gao S (2019) Marker-free coregistration of UAV and backpack LiDAR point clouds in forested areas. ISPRS J Photogramm Remote Sens 147:307–318
Raynolds MK, Walker DA, Maier HA (2006) NDVI patterns and phytomass distribution in the circumpolar Arctic. Remote Sens Environ 102(3–4):271–281
Sato H (2009) Simulation of the vegetation structure and function in a Malaysian tropical rain forest using the individual-based dynamic vegetation model SEIB-DGVM. For Ecol Manag 257(11):2277–2286
Srinivasan S, Popescu SC, Eriksson M, Sheridan RD, Ku N-W (2015) Terrestrial laser scanning as an effective tool to retrieve tree level height, crown width, and stem diameter. Remote Sens 7(2):1877–1896
Su Y, Guan H, Hu T and Guo Q (2018). The Integration of Uavand Backpack Lidar Systems for Forest Inventory. Paper presented at IGARSS 2018-2018 IEEE International Geoscience and Remote Sensing Symposium, IEEE, 8757-8760, July, 2018. Valencia, Spain.
Tao S, Wu F, Guo Q, Wang Y, Li W, Xue B, Hu X, Li P, Tian D, Li C (2015) Segmenting tree crowns from terrestrial and mobile LiDAR data by exploring ecological theories. ISPRS J Photogramm Remote Sens 110:66–76
Thomas SM, Chan Y-T (1989) A simple approach for the estimation of circular arc center and its radius. Comput Vision Graph Image Process 45(3):362–370
Wu J, Yao W, Choi S, Park T, Myneni RB (2015) A comparative study of predicting DBH and stem volume of individual trees in a temperate forest using airborne waveform LiDAR. IEEE Geosci Remote Sens Lett 12(11):2267–2271
Wulder MA, White JC, Andrew ME, Seitz NE, Coops NC (2009) Forest fragmentation, structure, and age characteristics as a legacy of forest management. Forest Ecol Manag 258:1938–1949
Xing J, Zheng C, Feng C, Zeng F (2017) Change of growth characters and carbon stocks in plantations of Pinus sylvestris var. mongolica in Saihanba, Hebei, China. Chin J Plan Ecol 41:840–849 (in Chinese)
Yu Y, Li J, Guan H, Wang C, Yu J (2015) Semiautomated extraction of street light poles from mobile LiDAR point-clouds. IEEE Transact Geosci Remote Sens 53(3):1374–1386
Yun T, Jiang K, Hou H, An F, Chen B, Jiang A, Li W, Xue L (2019) Rubber tree crown segmentation and property retrieval using ground-based mobile LiDAR after natural disturbances. Remote Sens 11(8):903
Zhao X, Guo Q, Su Y, Xue B (2016) Improved progressive TIN densification filtering algorithm for airborne LiDAR data in forested areas. ISPRS J Photogramm Remote Sens 117:79–91
Zhu Z, Piao S, Myneni RB, Huang M, Zeng Z, Canadell JG, Ciais P, Sitch S, Friedlingstein P, Arneth A, Cao CX, Cheng L, Kato E, Koven C, Li Y, Lian X, Liu YW, Liu RG, Mao JF, Pan YZ, Peng SS, Penuelas J, Poulter B, Pugh TAM, Stocker BD, Viovy N, Wang XH, Wang YP, Xiao ZQ, Yang H, Zaehle S, Zeng N (2016) Greening of the earth and its drivers. Nat Climate Change 6(8):791–796
We would like to thank Beijing Green Valley Technology Co. Ltd. (https://www.lidar360.com), providing Libackpack laser scanning system and LiDAR360 software for effectively manipulating LiDAR point cloud data.
This study is supported by the projects (41790425; 41971228) of Natural Science Foundation of China.
Ethics approval and consent to participate
The subject has no ethic risk.
Consent for publication
The authors declare that they have no competing interests.
About this article
Cite this article
Xie, Y., Zhang, J., Chen, X. et al. Accuracy assessment and error analysis for diameter at breast height measurement of trees obtained using a novel backpack LiDAR system. For. Ecosyst. 7, 33 (2020). https://doi.org/10.1186/s40663-020-00237-0
- Adaptive cylinder fitting
- Diameter at breast height
- Point cloud slice
- Point density
- Transitional model